- Dec 16, 2020
My apologies. I must’ve read it wrong. Thought you said it shifted 1 MOA or so after your testing, and then a few months later someone asked about the failure review and you said it was a dismounting/remounting issue. I don’t know where the thread was. But I read it in depth before I tested any March’s lol.I retracted nothing. The March lost zero. I won’t run one again. I don’t care if it’s due to weak internals or a weak tube not holding zero is a scope not worth using. I’ll repeat, the issue with tested with the scope attached is the rings move. Causation is not correlation. I can smack the rings and see reticle shift. This is another topic of discussion and one I’m still searching for permanent resolution.